The Master-Touch™ FAT™ probes utilize a flow averaging tube to give a stable flow signal in applications where the flow profile is less than ideal, such as downstream of a bend, valve, tee or obstruction. The flow averaging tube has a number of large diameter (0.125”) inlet ports along the length of the upstream impact surface. The impact pressure at each inlet port is averaged inside the tube to create the axial flow through the tube and across our flow sensor. The gas flow then passes back into the main flow stream through the gas return ports located near the flow sensing elements.
Anomalies in the actual flow profile or installations in non-circular ducts may require a C-Factor adjustment for the best accuracy. The velocity impact pressure follows a square root function. Therefore, the average velocity pressure in the FAT™ probe may vary slightly from the average of the velocities at each inlet port. The Master-Touch™ electronics easily accommodate the C-Factor adjustment.
The FAT technology is available in the same range of physical configurations as the other Eldridge Products' instruments. Inline style averaging tubes are available for pipe sizes 2" and greater with flanged ends; insertion style FAT probes can be as small as 2" or as large as necessary (90+" to date!). Integral and remote electronics with approvals for hazardous or non-hazardous locations allow the FATs to meet most installation requirements.
The Master-Touch™ MP Series Flow Averaging Tubes are approved by CSA/CUS, ATEX, IECEx and KOSHA for use in hazardous area locations.
The Master-Touch™ MPNH Series Flow Averaging Tubes are approved by CSA/CUS for use in general purpose (Ordinary) area locations.